The PosiTector Replica Tape Reader (RTR) is a hand-held electronic instrument that measures burnished TestexTM Press-O-FilmTM replica tape to produce peak-to-valley surface profile height measurements.
It consists of a body (Standard or Advanced) and probe. This digital spring micrometer measures and records peak to valley surface profile height using replica tape.
Replica Tape provides a simple way to obtain an impression of a surface for analysis. It consists of a layer of crushable plastic micro foam affixed to a 50.8 μm incompressible polyester film. When compressed against a roughened surface, the foam collapses and acquires an impression, or reverse replica, of the surface. It is available in a number of grades to accommodate measurements in different profile ranges.
Placing the compressed tape (replica) into the PosiTector RTR gives a measure of the average maximum peak-to-valley height of the surface roughness profile.
|Feature Comparison Chart||Standard
|Hi Contrast LCD Display||
|FAST mode - faster measurement speed for quick inspection|
|SCAN Mode - take continuous readings without lifting the probe||-||
|Hi-Lo limits audibly and visibly alerts when measurements exceed your specified limits||
|Instantly calculate average, standard deviation, min/max thickness and number of readings while measuring||
|Onscreen help, real time graphing, picture prompting and batch notes||-|
|Memory Features & Capacity||Standard
|Reading Storage Capacity||1000 in one group||250,000 in up to 1000 groups|
|Create meaningful batch names and enter notes directly on the instrument using a familiar on-screen QWERTY keyboard.||-|
|SSPC PA2 feature set determines if film thickness over a large area conforms to user-specified min/max levels||-|
|PSPC 90/10 feature determines if a coating system complies with an IMO performance standard for protective coatings||-|
|Multiple stored calibration adjustments for measuring on a variety of substrate conditions||-|
|USB port for fast, simple connection to a PC and to supply continuous power. USB cable included|
|WiFi technology wirelessly synchronizes with PosiSoft.net, downloads software updates, and connects to mobile devices for expanded functionality||-|
|Bluetooth Wireless Technology for data transfer to a PC or optional portable printer||-|
|PosiSoft Desktop Software Desktop software for PC or Mac computer|
|PosiSoft USB Drive Access your PosiTector as a flash drive. View and print readings and graphs using universal PC/Mac web browsers or file explorers|
|PosiSoft.net (formerly PosiTector.net) Cloud-based application offering centralized management of PosiTector readings|
|PosiSoft Mobile Gage-based software application featured in all PosiTector Advanced instruments||-|
|Measuring principle:||Digital spring|
|Display||Backlit & Reversible LCD|
|Data Output||IR & USB - "1" & "3" Models|
Bluetooth - "3" Models
|Measurement Range||20 to 115 µm|
|Accuracy (H)||± 5 µm|
|Resolution||± 1 µm|
|Anvil Diameter||⌀6.3 mm|
|Anvil Pressure||110 grams-force (1.1 Newtons)|
|Dimensions||152 mm x 61 mm x 28 mm|
|Weight||140 g without batteries|
PosiTector RTR Kit includes:
Prepare the test surface
Locate a representative site for measurement. Clean the surface to be tested.
We recommend the use of cleaning putty to remove dust, debris, or residual blast media from the surface.
Firmly press the putty onto the surface using your fingers, and remove.
Burnish the replica tape (create a replica)
Select the appropriate grade of replica tape based on the target profile (Coarse or X-Coarse). See tape instructions for assistance.
Pull a single adhesive-backed replica tape free of its release paper. A "bull's eye" circle of paper should remain on the release paper (it is not used for measurement).
Apply replica tape to blasted surface. Press the adhesive-backed ends of the paper to hold it firmly in place during the burnishing process.
Firmly compress replica film using the rounded end of the included stainless steel burnishing tool. Apply sufficient pressure in a circular and x-y rubbing motion to produce a replica with a uniform pebble grain appearance. It normally takes 30 to 60 seconds to fully compress all parts of the film on most surfaces. In general, too much compression is safer than too little.
After burnishing is complete, remove the replica tape from the surface.
Prepare the PT-RTR
Clean the measurement surfaces (anvils) using the included cleaning card. Swipe the card through the opening a few times while firmly pressing both probe buttons to remove dust particles and residual tape adhesive.
Power-up the PosiTector RTR by pressing the center navigation button. An image appears indicating that the probe requires zeroing.
Simultaneously press and hold both probe buttons firmly until the Gage beeps and the arrows point outward. Do not place anything in the measurement opening during this procedure.
Measure the included check shim to verify accuracy.
Ensure the correct grade of replica tape, C or XC, is displayed in the upper left corner of the LCD.
Insert the burnished replica tape into the measurement opening. Ensure the tape is properly positioned so that the burnished region is centered within the opening in the probe with the adhesive (sticky) side down.
To position the tape, move the Testex™ Press-O-Film™ replica tape to the back of the probe and align the two dots (printed on replica tape) with arrows on both sides of the measurement opening.
If you are using older Testex™ Press-O-Film™ replica tape that does not have the printed dots, simply align by centering the tape within the measurement opening.
Once positioned, firmly press both probe buttons simultaneously and hold until the PosiTector RTR beeps and the measurement is displayed. During measurement, a constant anvil pressure is applied to the replica tape regardless of how hard the two buttons are pressed.
The PosiTector RTR automatically subtracts the 50.8 µm (2 mil) of incompressible polyester film. No further adjustments are required.
Replica Tape provides a simple way to obtain an impression of a surface for analysis. It consists of a layer of crushable plastic micro foam affixed to a 50.8 µm (2 mil) incompressible polyester film. When compressed against a roughened surface, the foam collapses and acquires an impression, or reverse replica, of the surface. It is available in a number of grades to accommodate measurements in different profile ranges.
Placing the compressed tape (replica) into the PT-RTR gives a measure of the average maximum peak-to-valley height of the surface roughness profile.
The gage automatically subtracts the thickness of the polyester substrate from all measurements.